The method is, in essence, the colorimetric method described by Strickland and Parsons (1968) as modified by Presley (1971) for DSDP pore fluids.
It is important to note that the concentration in the final test solution cannot exceed more than about 10 mM. Thus, for open-ocean (low sedimentation rate, low organic carbon) samples, one might need to do the determination on 2 cm3 of sample (expected range 0-10 mM), but in typical continental-margin settings, where concentrations can exceed 100 or 200 mM, a 0.1 or 0.2 cm3 sample aliquot must be used. As mentioned above, the concentration range must be established prior to running the samples, and it is highly advisable to make standards that cover the range of concentrations to be expected. In this manner, standards and samples will all get the same treatment.
The method is based on an adaptation of the method proposed by Strickland and Parsons (1968). In the method, nitrite is allowed to react with sulfanilamide in an acid solution. The resulting diazo-compound reacts with N-(1-naphtyl)-ethylene diamine to form a pink azo dye, whose absorbance is measured at 543 nm.
After 10 minutes, but before 2 hours has elapsed, measure at 543 nm in the 1 cm path length, flow-through cell. Use nanopure water as a blank.
Often a good judgment can be made for the potential use of the nitrate method on the basis of the ammonia measurements. If the latter rise very quickly above 50 mM, it is almost certain that little or no nitrate will be present but rather that the zone of sulfate reduction has been entered.
The method is adopted from Strickland and Parsons (1968) and makes use of the catalytic reduction of nitrate to nitrite, using a cadmium reduction column. A peristaltic pump (of autoanalyzer type) is used to force the samples and standards through the reduction columns. The use of only one channel of the pump is advocated to keep better track of the samples and standards. It should be remembered that each and every one of the columns has its own individual characteristics.
1. Teflon tubing of 3 mm internal diameter. Put a small amount of glass wool in the bottom of the teflon tube (fine copper wool is supposed to be preferable). Fill about 5 cm length of the tubing with small (0.5 to 2 mm) Cd chips. Put a small amount of glass or copper wool on top of the loosely packed column.
2. A more preferred method is to use < 1 mm (i.d.) tygon tubing and fill this with cadmium wire. A thin piece of copper wire can be used on both sides of the cadmium wire.
Of importance is to note that the columns, after their activation as described below, remain out of contact with air because they will get poisoned. Also avoid contact with hydrogen sulfide-containing solutions. They will produce CdS and finish the columns.
The columns get attached to 1/16-inch-by-1/8-inch tygon tubing which can be used in the peristaltic pump or at the other end. When not in use, keep both ends in water to prevent aeration of the columns. Intake------peristaltic pump------column------outflow.
Activation of the column(s) can be achieved as follows. (If starting with a new, clean Cd column, step 1 may be eliminated.)
1. Pass 5% HCl through the columns for a few minutes, then wash with
nanopure water until the effluent has a neutral pH.
2. Pass a 2% copper sulfate solution through the column for a few minutes (10-20 cm3), followed by a wash with dilute ammonium chloride (see below). The column is now ready.
Before running samples through the column, do a pre-rinse using dilute NH4Cl.
Use 1 cm3 of sample or standard and add 4 cm3 of nanopure water. Add 0.1 cm3 of concentrated NH4Cl solution as a buffer. Run the buffered sample through the column at a speed of ~3-5 cm3/minute. Collect the last 4 cm3. For the actual analysis of the nitrite, only 2 cm3 is used, as described above in the nitrite method.
After all the samples have been run through the column, a wash with dilute NH4Cl is advisable for the duration of at least several minutes. Also make sure that the intake tube and the outlet tube remain submersed in water in order to prevent any inadvertent contamination of the reduction column.
Standards are made from a stock solution of 10 mM KNO3. The range of the standards should be 0-60 mM. It is best to prepare the standards in synthetic seawater: 30 g NaCl, 10 g MgSO4.7H2O, 0.05 g NaHCO3 in 1000 cm3 of nanopure water.
Bates, R. G., and Culberson, C. H., 1977. Hydrogen ions and the thermodynamic state of marine systems. In Anderson, N. R., and Malahoff, A. (Eds.), The Fate of Fossil Fuel CO2 in the Oceans: New York (Plenum Press), 45-61.
Brumsack, H., Notes on Atomic Absorption, Leg 127 (in press).
Gieskes, J. M., and Lawrence, J. R., 1976. Interstitial water studies, Leg 35. In Hollister, C. D., Craddock, C., et al., Init. Repts. DSDP, 35: Washington, DC (U.S. Govt. Printing Office), 407-424.
Gieskes, J. M., and Rogers, C., 1973. Alkalinity determination in interstitial waters of marine sediments. J. Sed. Petrol., 43:272-277.
Gieskes, J. M., and Peretsman, G., 1986. Water chemistry procedures aboard JOIDES Resolution--some comments. ODP Technical Note 5: College Station, TX (Ocean Drilling Program).
Gieskes, J. M., Elderfield, H., Lawrence, J. R., Johnson, J., Meyers, B., and Campbell, A. C., 1982. Geochemistry of interstitial waters and sediments, Leg 64, Gulf of California. In Curray, J. R., Moore, D. G., et al., Init. Repts. DSDP, 64(2): Washington, DC (U.S. Govt. Printing Office), 675-694.
Grinstead, R. R., and Snider, S., 1967. Modification of the Curcumin method for low level boron determinations. Analyst, 92:532-533.
Kastner, M., Elderfield, H., et al., 1990. Diagenesis and interstitial water chemistry at the Peruvian Margin--major constituents and strontium isotopes. In Suess, E., von Huene, R., et al., Proc. ODP, Sci. Results, 112: College Station, TX (Ocean Drilling Program), 413-440.
Khoo, K. H., Ramette, R. W., Culberson, C. H., and Bates, R. G., 1977. Determination of hydrogen ion concentrations in sea water from 5 to 40C: standard potentials at salinities from 20 to 45%. Analytical Chem., 49:29-33.
Kremling, K., 1983. Determination of major constituents: bromide. In Grasshoff, K., Ehrhard, M., and Kremling, K. (Eds.), Methods of Sea Water Analysis. New York (Springer Verlag).
Pedersen, T. F., 1979. The geochemistry of sediments of the Panama Basin, eastern equatorial Pacific Ocean. Ph.D. Thesis, University of Edinburgh.
Presley, B. J., 1971. Techniques for analyzing interstitial water samples. Appendix Part 1: Determination of selected minor and major inorganic constituents. In Winterer, E. L., et al., Init. Repts. DSDP, 7(2): Washington, DC (U.S. Govt. Printing Office), 1749-1755.
Solorzano, L., 1969. Determination of ammonia in natural waters by phenol-hypochlorite method. Limnol. Oceanogr., 14:799-801.
Strickland, J.D.H., and Parsons, T. R., 1968. A manual for sea water analysis. Bull. Fish. Res. Bd. Canada, 167.
Tsunogai, S., Nishimura, M., and Nakaya, S., 1968. Complexometric titration of calcium in the presence of large amounts of magnesium. Talanta, 15:385-390.
Weiss, J., 1986. Handbook of Ion Chromatography. DIONEX Corp.
Return to Part I
Return to Part II